Easy to Use: Simply immerse the pH meter electrode in the solution to be tested, readings are quickly displayed on a large LCD display
Wide Application: Ideal instrument for any aquarium, the fishing industry, swimming pools, spa, school laboratories, food & beverage, drinking water, etc.
Auto Calibrated: Features in-built calibration with 2 buffer powders. Just press a button to finish the calibration. This pH meter is factory calibrated. Users can use it directly.
Fast and Accurate Results: No need to work with those annoying color charts and test strips.Resolution: 0.01 pH, Accuracy: ±0.01 pH
Compact Portable Design: Pocket fitting pH meter that’s also lightweight and compact. Total concealed fit in your bag or pocket for easy, casual carrying.
Display: LCD Digital
Operating Temperature: 0~60°C
Measuring Range: 0.00~14.00
Probe Type: Precision glass bulb electrode
Power source: 2 x 1.5 V(LR44 Button Cell) (not included)
Step 1.Remove protective cap.
Step 2.First rinse the electrode with distilled water and suck it with filter paper.
Step 3.Turn the meter on by pressing the “ON/OFF” key.
Step 4.Immerse the pH meter electrode in the solution to be tested (should not be over the immersion line).
Step 5. Stir gently and wait around 30 seconds till the reading stabilizes.
Step 6. After finishing, clear the electrode with pure water, turn the meter off by pressing the “ON/OFF” key.
Step 7. Always replace the protective cap after use.
1. Turn on PH meter
2.Dissolve each buffer solution in 250ML of distilled water
3.Immerse the electrode into the PH 6.86 solution,(under the temperature of 25)
4.Press the “CAL”(calibration)button for 5 seconds and release
-The display will start flashing 6.86
-Wait until the display stops flashing.
Rinse the electrode with distilled water and dry it with filter paper.
5. Immerse the electrode in PH 4.00 solution,
-Press“CAL” button for 5 seconds, then press and release immediately the second time
-Reading will start flashing 4.00
-Wait until the display stops flashing
Rinse and dry the electrode with distilled water as before.
6. Place the meter back to the buffer solution to test calibration.If incorrect, please repeat the calibration
7. If the approximate PH of your test solution is known to be above 7.0, calibrate the meter using 6.86 and 9.18 solution. To do this, please follow the calibration chart, The third step is the calibration method for 9.18 points.
1) 1pcs pH Meter
2) 2 Bags Buffer Powder(4.00/4.01,6.86)
3) Instruction Manual
Get rewards while shopping for your gadgets. Join now to start collecting Cove Coins 🤩